We have 6 Fresenius Medical Care K manuals available for free PDF download: Troubleshooting Manual, Operator’s Manual, Technical Manual. View and Download Fresenius Medical Care K operator’s manual online. K Hemodialysis Machine Operator’s Manual. K pdf manual download. FRESENIUS K OPERATORS MANUAL KHOXJMBJNW | PDF | 45 Pages | KB | 06 Apr FRESENIUS K OPERATORS MANUAL.
|Published (Last):||28 April 2018|
|PDF File Size:||10.2 Mb|
|ePub File Size:||11.39 Mb|
|Price:||Free* [*Free Regsitration Required]|
These instructions are for the Fresenius K Hemodialysis machine. Observe the system for fluid leaks and discontinue treatment if you are unable mqnual correct any fluid leak quickly.
This program is used to remove excessive air from the hydraulic system. The Adult size comes standard with the K and can accommodate patients with upper arm circumferences of centimeters. An X will appear in the On check box next to the button when the option is selected.
Fresenius Medical Care K Manuals
By pressing ‘print’ button you will print only current page. Arterial Pressure Calibration Test Check UF pump. The machine is in bypass mode during the test period.
It cant take the FDA 15 years! Manufacturer’s Electromagnetic Compatibility emc Declaration The Standard Prime method allows the operator to prime the blood circuit by controlling the flow of the saline manually. The K can be configured so that this testing is mandatory after power up providing that the force test option is selected.
General Warnings General Warnings This section contains general warnings statements regarding the use and maintenance of the Fresenius K hemodialysis machine.
Fill Program—Occurs when water level in the air separation chamber gets too low. Forced Test Upon power up and entering dialysis except after a power failure or short power downthe machine will be in Standby. Heater is off during pre-rinse. Section 10 – Pressure Tests Failing Be sure freseniux open the door to the optical detector before pulling the line from the venous clamp and optical detector assembly.
Arterial, Venous And Transmembrane Pressure Manuual machine should be rinsed thoroughly after chemical disinfection and before introducing any other chemicals be rinsed thoroughly after chemical disinfection and before introducing any other chemicals to the machine. The performance of bloodlines not manufactured by Fresenius Medical Care cannot be guaranteed by Fresenius and are therefore the responsibility of the prescribing physician. BVM will no longer pass information to the Freaenius monitor.
Venous Drip Holds the drip chamber and aligns it with the ultrasonic air sensor. Alarms And Troubleshooting Testing Concentrate And Bicarbonate Pumps Page 32 The dialysate delivery machine can be used with a variety of blood-tubing configurations.
Fresenius Medical Care 2008K Manuals
Table of contents Troubleshooting Electronic Circuit Description Arterial Chamber This option is used to define whether the arterial chamber is pre-pump or post-pump.
The K hydraulic system is a closed system that uses a separate UF pump for greater accuracy. Temperature Control Calibration method 1 These screens are the: If the failure message is the automated Test Sequence has repeated on retest, take the machine out of service and alert a failed. I am just looking at the machine to use as a study guide for School. Home Screen Monitoring