D CV. Read this owner’s manual thoroughly before use Owner’s operation manual for series. DT Series. 3 1/2 Digital Multimeter. Read this owner’s manual thoroughly before use. DT series multimeter have been designed according to IEC concerning electronic measuring. View and Download Histest DT owner’s operation manual online. DT Multimeter pdf manual download.
|Published (Last):||5 June 2014|
|PDF File Size:||9.69 Mb|
|ePub File Size:||11.99 Mb|
|Price:||Free* [*Free Regsitration Required]|
Before attempting to open the case, always be sure that test leads have been disconnected from measurement mqnual. The voltage is approximate 5Vp-p. Operating Instruction “COM” jack.
HISTEST DT OWNER’S OPERATION MANUAL Pdf Download.
By pressing ‘print’ button you will print only current page. Use the proper terminals, function, and range for your measurements. Set the range switch to the desired range.
Set the range switch to ” ” or ” ” range. If the resistor to be kanual is connected to a circuit, disconnect circuit power and discharge all capacitors before measuring resistance. Close case and tighten screws completely before using the meter to avoid electrical shock hazard.
Histest DT830 Owner’s Operation Manual
Safety Information If the defect has been caused by a misuse or abnormal operation conditions, the repair will be billed at a nominal cost. To print the manual completely, please, download it.
Comments to this Manuals Your Name. Enter text from picture: Don’t show me this message again.
Page 6 To use safely, each measurement can not last for more than 10 seconds, and the interval between each measurement must be more than manuzl minutes. If the defect has been caused by a misuse or abnormal operation conditions, the repair will be billed at a nominal cost. Front Panel Description This function allows the meter to output a signal as a signal generator.
This function nanual the meter to output a signal as a signal generator. Don’t have an account? Got it, continue to print.
Owner’s operation manual for DT Series 3 ½ Digital Multimeter
A coupling capacitor should be used when connecting the meter to a circuit. Table of Contents Add to my manuals Add. In the latter case, the transient stresses are variable; for that reason, its necessary that the transient-withstand -capability of equipment is made known to the user.
Disconnect circuit power and discharge all high-voltage capacitors before testing resistance, continuity, diodes, or capacitance. Battery And Fuse Replacement Before attempting to open the case, always be sure that test leads have been disconnected from measurement circuits.
Related Manuals for histest DT No related manuals.
To use safely, each measurement can not last for more than 10 seconds, and the interval between each measurement must be more than 15 minutes. Before measuring current, check the meter’s fuses and turn power off to the circuit before connecting the meter to the circuit. A test signal will appear between the and manial jacks. Page of 13 Go.