ASTM D is a test method used to study the compressive properties of a composite laminate that has a hole cut into it. The GD fixture meets the requirements of ASTM D for open-hole compressive strength testing of fiber-reinforced polymer-matrix composite. The ASTM D / DM fixture is used to perform open-hole compressive tests on multi-directional polymer matrix composite laminates reinforced with high .
|Published (Last):||3 March 2008|
|PDF File Size:||10.61 Mb|
|ePub File Size:||15.81 Mb|
|Price:||Free* [*Free Regsitration Required]|
If the gap is too large, shim between the support plate and the long grip, holding the shim stock on the support plate with tape. Factors that influence the notched compressive strength and shall therefore be reported include the following: A superscript epsilon e indicates an editorial change since the last c6484 or reapproval. The force is transmitted by shear into the support? Two acceptable test procedures are provided. This device shall be essentially free from inertia lag at the speci?
While the hole causes a stress concentration and reduced net section, it is common aerospace practice to develop notched design allowable strengths based on gross section stress to account for various stress concentrations fastener holes, free edges,?
ASTM D6484 Testing Fixture
Retests shall be performed for any specimen on d684 values are not calculated. The torque level may be increased above this level when acceptable failure modes cannot be achieved using 7 N-m [60 in-lbf] applied torque.
Hydraulic grip-loaded data typically exhibit linear behavior at the onset of loading. An optional threaded support plate is shown in Figs.
ASTM D Testing Fixture
Threeplace failure mode descriptors, summarized in Table 1, shall be used. The ends of the grip-jaws on wedge-type grips should be even with each other following insertion to 11 avoid inducing a bending moment which could result in wstm failure of the specimen.
However, since the notch dominates the strength, consistent preparation of the hole, without damage to the laminate, is important to meaningful results. In laminated composites, the principal material coordinate system has meaning only with respect to an individual orthotropic lamina. Terminology Aastm and Practice E de? Summary of Test Method 4.
The following general notes apply to these? NOTE 2—Experience has shown that? The long and short? Results may also be affected by the ratio of ungripped specimen length to specimen width; this ratio should be maintained at 2.
Sources of Additional Information: Historical Version s – view previous versions of standard.
The test laminate con? If the specimen is to be failed, record the maximum force, the failure force, and the head displacement at, or as near as possible to, the moment of rupture. If you feel that your comments have not received a fair hearing you should make your views known to the ASTM Committee on Standards, at the address shown below.
If the specimen is end-loaded, it may be desirable to d64484 some additional clamping force at the ends of the specimen, to minimize end-brooming. Originally approved in Within the text the inch-pound units are shown in brackets.
Report the method of sampling. Factors considered should include the? Holes should be drilled undersized and reamed to? At astmm force levels, some nonlinear behavior may d6844 observed due to grip slippage.
ASTM D Compression Fixture for Fiber Composite Laminates| GD
If the d66484 strength of the material cannot be reasonably estimated, initial trials should be conducted using standard speeds until the ultimate strength of the material and the compliance of the system are known, and speed of testing can be adjusted. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
If the platens are not sufficiently hardened, or simply to protect the platen surfaces, a hardened plate with parallel surfaces can be inserted between each end of the?